How to clean the NObreath properly
Bedfont® recommends wiping the instrument’s external surfaces with a product specifically developed for this purpose. It is recommended that wipes are used once and for one surface only.
The NObreath® device should be cleaned for initial use and after each patient use. Please do not use alcohol, cleaning agents containing alcohol or other organic solvents as these vapours will damage the electrochemical sensor inside.
If the problem persists or has not been resolved by the information provided above, please contact Bedfont or your local distributor for more advice.