Why will the Gastro+ not turn on?
There may be a couple of reasons why the Gastro+ Gastrolyzer will not turn on:
- This may simply be an issue when the batteries are empty. First, you should replace the batteries and check if this resolves the problem.
- It could be that the programming switch hasn't been set to 0 after a software update. To resolve this, remove the battery cover and set the switch to 1.
- Another reason the Gastro+ monitor might not turn on is if the screen driver has not been updated after a screen change or the firmware has been corrupted. Both of these issues can be fixed by reprogramming the device via the Bedfont Monitor Recovery Tool.
- If the batteries have leaked it may have damaged the monitor. You should contact your distributor to arrange a PCB repair/replacement.
If the problem persists or has not been resolved by the information provided above, please contact Bedfont or your local distributor for more advice.