Why will my 1st generation NObreath not switch on?
- If your 1st generation NObreath doesn't seem to be turning on, please try replacing the batteries in the first instance.
- If the monitor is still not switching on, is is possible that the wrong screen has been used when being repaired? See below the correct screens you need:
- DIS024 – v2.3
- DIS037 – v2.4
- DIS037 – v3.1
- If it is none of the above, it could either be that the PCB has failed electronically or the battery or sensor has leaked. In this instance, please contact your local distributor or Bedfont to arrange a repair.
If the problem persists or has not been resolved by the information provided above, please contact Bedfont or your local distributor for more advice.